New Dental X-ray Film Automatic Processor Developer
- The oral automatic X-ray film bath equipment is a total automatic and wall mounted type.
- The film is bathed automatically through three step: Developing, fixing and bathing.
- The time developing films needed can be selected in the designing range. During the bathing process, delivery mechanism adopts anti-flaming ABS engineering materials.
- It has easy cleanliness and maintenance, stable running, small volume, operators do not touch the developers and fixers directly, convenient and safe operation and so on characteristics.
Main Technical parameters:
- Normal working conditions
- A). Surroundings temperature: 18º C-30º C;
- B). Relative humidity:≤70%;
- C). Atmospheric pressure: 700hPa-1060hPa;
- D). Voltage of electrical source: 110V~ 220V± 10V; (this can be used worldwide,i will send the the correct one to fit your country.)
- E). Frequency of electrical source: 50Hz± 1Hz.
Surroundings of transportation and reservation:
- A). Surroundings temperature range: -40º C-+55º C;
- B). Relative humidity range: ≤93%
- C). Atmospheric pressure range: 500hPa-1060hPa.
- (1). Processing time of film
- The time which develops films needed is 2min-8min, and can be accommodated continuously. The error is within± 10%.
- (2). Light leaking
- During the bathing process, light leaking is not permitted.
- (3)The stability of film delivery
- The delivery mechanism of film should not lacerate or lock films.
- (4)The timing panel of transit film
- The timing panel of transit film adopts the number 1-12 represents month, each scale represents 10 days.